Nikon Europe B.V. and all Nikon subsidiaries in Europe are closely monitoring the situation with COVID-19 (coronavirus) and following the guidance of each local health authorities. Learn more about the measures we have put in place.
The connection between your microscope and camera.
There are literally hundreds of potential combinations of camera and microscope types. The most fundamental information when deciding which c-mount adapter needed are as follows:
What make and model is your microscope?
This information is important as there will be different focal positions (point at which image data will be collected on the cameras chip) for different manufacturers microscopes. Modern microscopes will tend to have the model type on the microscope body, sometimes near the serial number. As an alternative, the microscope manufacturer should be able to identify the microscope by its description or by a photograph.
What make and model is the camera you are trying to attach?
This is an important piece of information as it usually determines the type of thread or fitting the c-mount adapter should have.
What port will you be using?
It is not always possible to connect a camera via a dedicated camera port (sometimes referred to as trinocular tube or beamsplitter). It is important that you identify which port will be used to accept the c-mount.
Is a magnification necessary?
The camera chip (CCD) will be of a certain dimension and will require the appropriate magnification within the c-mount adapter (sometimes referred to as a relay lens.)
A good rule of thumb is the size the CCD as a decimal is what is required in magnification to fill the viewfield. For example, a 2/3” CCD (0.67”) requires a 0.7x c-mount. The CCD size should always be made available by the manufacturer.